Ruijin Cordless Battery Autoclavable Orthopedics Economic Bone Drill
|Purchase Qty. (Sets)||FOB Price|
|1-29||US $800 / Set|
|30+||US $400 / Set|
I; Portable and lightweight
Ii: Reliable and performance
2: Technical Data
Hand Piece Working Voltage 14.4V
Hand Piece Output Power ≥ 20W
Hand Piece Unit Rise In Temperaturer ≤ 25° C
Charger Input Voltage AC 110v-240V/ 50Hz-60Hz
Charger Output Power 14.4V, 900mA
Charger Input Power 30VA
Battery Voltage 14.4V
Battery Charging Time About Three Hours
3: Standard Configuration
I: One hand piece
Ii: One battery
Iii: One Charger
Iv: One drill chuck
V: One Sterilizing channel
Vi: One Aluminium box
4: Technical Merits
I: Axial run-out of cincentricity is within 0.05mm
Ii: Rotating speed: 600 rpm
Iii: Ritated torque 3.5N/m
Iv: Supporting the interface of the brand productsv: Semiluser surface treatment technology gives the operator more comfortable operation
Vi: Multipurpose interface is easier to use in clinic
Vii: 4.0mm hollow for all brand needles to pass througth
Viii: Ni-MH battery without memory environment
5: Payment Terms
I: Price: FOB USD400-800 / Set, Shanghai Port
Ii: MOQ: 1 piece
Iii: Payment Terms: T/T in advance or western union
Iv: Delivery Time: Within 3 days after payment
1. Sterilize temperature ≤ 135 ° C, if the temperature is over 135 ° C, the motor will be damaged.
2. Normal bone drill can only hold drill bits less than 6mm, acetabulum reaming drill with big torque,
Rpm is around 300, canulate drill is used with nail or K wire, rpm is around 600.
3. The battery need to be get out of the power tool after surgery. Otherwise the small automatic
Discharge of the electric will make the life of the battery short.
4. The battery cannot be sterilized, it is not autoclavable.
5. The cranial bur must be washed right after the surgery by purger in order to avoid the blood
Stick in the bur, then we should make it dry.
6. The battery space in the handpiece of the tool should not have any water, after sterilization.
7. The cranial bur blade is made of medical stainless steel, in order to avoid broken, so it is flexible
And not very hard, it is easy to be attrite, we suggest each blade only use for 1 surgery.
8. The medical electric drill, saw only the equivalent of medical doctors or professional use in
A short time, can not use a long time continuously.
9. Charger power cord should not be traces of oil, solvents and other chemicals, erosion or mechanical
Damage, if it is found to be immediately replaced.
10. Users may not disassemble or repair, or responsibility by the user, the Company is not responsible for
Repairs. Man-made damage is not covered under warranty, the costs is incurred by the users themselves.
11. The tools need repairing during the warranty period, should be packaged well before transportation.
|Output Power||95 W|
|Unit rise in temperature||≤ 25 °C|