CE Approved Rechargeable Battery Operated Medical Autoclave Multifunction Bone Drill Saw
|Purchase Qty. (Pieces)||FOB Price|
|1-29||US $4,000 / Piece|
|30+||US $2,500 / Piece|
Electric surgical multifunction drill&saw system is one of power tools widely used in orthopedic, cranial, thoracic, joint replacement operation.
Gradually replaced traditional industrial drill and saw, it can be autoclaved up to 135° C high temperature and 1.6Mpa high pressure. Meanwhile, the different attachment can change by each other into the same hand piece. It will be easier for surgery to choose different attachment when do different operation.
1. Our whole power tool can be autoclavable.
2. We can autoclavable for 135 degree, with 1.6Mpa, in this way the bacilli and virus which can
Not killed by the suffocating will be killed.
3. We use nickel-metal hydride battery is safe in use.
4. The machine has very low noise.
5. Our products has 1 year warranty period of hand piece, has 3months quality warranty of the article of consumption.
1. Sterilize temperature ≤ 135 ° C, if the temperature is over 135 ° C, the motor will be damaged.
2. Normal bone drill can only hold drill bits less than 6mm, acetabulum reaming drill with big torque,
Rpm is around 300, canulate drill is used with nail or K wire, rpm is around 600.
3. The battery need to be get out of the power tool after surgery. Otherwise the small automatic
Discharge of the electric will make the life of the battery short.
4. The battery cannot be sterilized, it is not autoclavable.
5. The cranial bur must be washed right after the surgery by purger in order to avoid the blood
Stick in the bur, then we should make it dry.
6. The battery space in the handpiece of the tool should not have any water, after sterilization.
7. The cranial bur blade is made of medical stainless steel, in order to avoid broken, so it is flexible
And not very hard, it is easy to be attrite, we suggest each blade only use for 1 surgery.
8. The medical electric drill, saw only the equivalent of medical doctors or professional use in
A short time, can not use a long time continuously.
9. Charger power cord should not be traces of oil, solvents and other chemicals, erosion or mechanical
Damage, if it is found to be immediately replaced.
10. Users may not disassemble or repair, or responsibility by the user, the Company is not responsible for
Repairs. Man-made damage is not covered under warranty, the costs is incurred by the users themselves.
11. The tools need repairing during the warranty period, should be packaged well before transportation.
1. RJ0610 hand piece
2. Two batteries
3. One charger
4. Two sterilization channel
5. RJ2002 quick chuck attachment (option)
6. RJ2001 oscillating saw attachment (option)
7. RJ2003 reamer attachment (option)
8. RJ2005 wire and pin attachment (option)
9. RJ2008 sternum saw attachment (option)
10. RJ2009 craniotomy mill attachment (option)
11. RJ2007 cranial drill attachment (option)
12. Aluminum box/Carry Case
13. High temperture cover
Additional battery is chargeable
The handpiece guarantees for one year, the battery and other fast consumption items guarantee
For 3 months.
|Bone drill speed||0-1300r/m|
|Cannulated drill speed||0-600r/m|
|Acetabular bone drill||0-300r/m|
|Saw non-load noise||≤70dB|
|Drill non-load noise||≤60dB|